{
   "crc": "DEADBEEF",
   "crc-root": "payload",
   "payload": {
      "header": {
         "fileType": "ProfileLibrary",
         "schemaVersion": "1.1",
         "fileVersion": "5.3.13",
         "releaseDate": "2026-06-25"
      },
      "items": [
         {
            "type": "category",
            "id": "c-1",
            "label": "Modality",
            "order": "1",
            "items": [
               {
                  "type": "folder",
                  "id": "f-2",
                  "label": "Radiography",
                  "order": "1",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-21",
                        "label": "W/Al AGMS Diag",
                        "labelLong": "W-Al diagnostic machines AGMS Diag",
                        "profile": "AGMS1_STD_10.agp",
                        "sensor": "AGMS",
                        "order": 1,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: W/Al, General 40-150 kV, 2-40 mm Al",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-40",
                        "label": "AGMS Diag W/Al - ROI last 0.1s",
                        "labelLong": "AGMS Diag W/Al with ROI for last 100 ms",
                        "profile": "AGMS1_std_adj0.1s_10.agp",
                        "sensor": "AGMS",
                        "order": 2,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "extractedRegion": "0.1 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: W/Al, AGMS, provides 0.1 s ROI at end of exposure, 40-150 kV, 2-40 mm Al",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-86",
                        "label": "AGMS Diag - ROI last 0.1s + Ich",
                        "labelLong": "AGMS Diag W/Al with ROI for last 100 ms + Ich",
                        "profile": "AGMS1_std_icf_adj0.1s_10.agp",
                        "sensor": "AGMS/ICH",
                        "order": 3,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "extractedRegion": "0.1 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: AGMS + ion chamber with 100 ms ROI at end of exposure, W/Al 40-160 kV, 2-30 mm Al",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-42",
                        "label": "DD Std trig + Ich",
                        "labelLong": "Dose Diode Std trigger + Ion Chamber",
                        "profile": "DD_std_icf_10.agp",
                        "sensor": "DD/IC",
                        "order": 4,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "0",
                           "kvMax": "0",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "General Radiation: Use a dose diode to trigger low dose rate ion chamber measurments. ",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-36",
                        "label": "10X6-60DAP",
                        "labelLong": "10X6-60DAP as DAP chamber to measure DAP",
                        "profile": "icf_std_dap_10.agp",
                        "sensor": "DAP",
                        "order": 5,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "DAP: 10X6-60DAP, Use chamber positioner & expose within the outline, 40-150 kV, 1.5-40 mm Al.",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-78",
                        "label": "10X6-60DAP as Ion Chamber",
                        "labelLong": "10X6-60DAP as Ion Chamber to measure Dose",
                        "profile": "icf_std_10.agp",
                        "sensor": "IC",
                        "order": 6,
                        "meta": {
                           "manufacturer": "Radcal",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: 10X6-60 DAP measure Dose.W/Al, 40-150 2-40 mm Al,  Expect wider energy response tolerance, +5%/-10%",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-3",
                  "label": "Fluoroscopy",
                  "order": "2",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-33",
                        "label": "Fluoro AGMS Diag adj - 1s ROI",
                        "labelLong": "Continuous Fluoro for adjusting with last one second region AGMS Diag",
                        "profile": "agms1_low_adj1.0s_10.agp",
                        "sensor": "AGMS",
                        "order": 1,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Fluoro",
                           "extractedRegion": "1.0 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Fluoro: AGMS, W/Al General, - provides 1s ROI values at end of exposure, 40-150 kV, 2-20 mm Al,  Exposure> 1s., Low threshold",
                           "baseCal": "WalDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "LOW"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-60",
                        "label": "Contin. Fluoro Ich - 1sec ROI",
                        "labelLong": "Continuous Fluoro using Ion Chamber for adjusting with last one second region",
                        "profile": "icf_low_adj1.0s_10.agp",
                        "sensor": "ICH",
                        "order": 2,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Fluoro",
                           "extractedRegion": "1.0 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Fluoro (not pulsed): Ion Chamber, when adjusting at very low dose rates.ROI for last 1 second.  W/Al 40-160kV",
                           "baseCal": "WalDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "LOW"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-41",
                        "label": "Fluoro AGMS dose only adj - 1s",
                        "labelLong": "Continuous Fluoro using AGMS Diag for adjusting at low dose with last one second region",
                        "profile": "AGMS1_low_DoseOnlyadj1.0s_10.agp",
                        "sensor": "AGMS",
                        "order": 3,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Fluoro",
                           "extractedRegion": "1.0 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": true,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Fluoro (not pulsed): AGMS, W/A,l dose & dose rate only, no kV, For long fluoro runs when kV signal is too low. Last 1s ROI provides dose & dose rate, 40-150 kV, 2-40 mm Al, Dose accuracy = 10%. Low threshold.",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "LOW"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-61",
                        "label": "Pulsed Fluoro AGMS adj -1s ROI",
                        "labelLong": "Pulsed Fluoro using AGMS Diag for adjusting with last one second region",
                        "profile": "agms1_std_adj1.0s_10.agp",
                        "sensor": "AGMS",
                        "order": 4,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Fluoro",
                           "extractedRegion": "1.0 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "150",
                           "kvMax": "40",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": true,
                           "tube": "General",
                           "conditions": "Fluoro (pulsed): AGMS, for adjusting at very low dose rates, Dose only, no kV. ROi for last 1 s.W/Al, 40-150 kV, 2-40 mm Al",
                           "baseCal": "WalDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-4",
                  "label": "CT",
                  "order": "3",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-29",
                        "label": "CT W/Al AGMS Diag",
                        "labelLong": "CT machines W/Al AGMS Diag",
                        "profile": "AGMS1_STD_GeneralCT_10.agp",
                        "sensor": "AGMS",
                        "order": 1,
                        "meta": {
                           "manufacturer": "General",
                           "model": "CT",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "120",
                           "kvMax": "140",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "1.35",
                           "paddle": false,
                           "tube": "General CT",
                           "conditions": "CT: W/Al, AGMS, tube stationary @0\u00B0, 120-140kV, 2-40 mm Al, Bow-tie filter out, service mode ",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-81",
                        "label": "Ich Slow rise low trigger",
                        "labelLong": "Ion Chamber - Low trigger, filtered",
                        "profile": "ics_low_10.agp",
                        "sensor": "IC",
                        "order": 2,
                        "meta": {
                           "manufacturer": "Radcal",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "0",
                           "kvMax": "0",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: W/Al Any ion chamber, low trigger, filtered  (Slow rise time for improved noise rejection).",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "LOW"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-94",
                        "label": "Ich Slow rise std trigger",
                        "labelLong": "Ion Chamber - Std trigger, filtered",
                        "profile": "ics_std_10.agp",
                        "sensor": "IC",
                        "order": 3,
                        "meta": {
                           "manufacturer": "Radcal",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "0",
                           "kvMax": "0",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: W/Al Any ion chamber, std trigger, filtered   (Slow rise time for improved noise rejection).",
                           "baseCalId": "0",
                           "dateLastMod": "2024-04-26",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-95",
                        "label": "DD Std trig + Ich slow",
                        "labelLong": "Dose Diode Std trigger + Ion Chamber, filtered",
                        "profile": "DD_std_ics_10.agp",
                        "sensor": "DD/IC",
                        "order": 4,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "0",
                           "kvMax": "0",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "General Radiation: Use a dose diode to trigger low dose rate ion chamber measurements  (slow filter for improved noise rejection).",
                           "baseCalId": "0",
                           "dateLastMod": "2024-08-15",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-5",
                  "label": "Mammography",
                  "order": "4",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-4",
                        "label": "General Mo/Mo AGMS Mammo",
                        "labelLong": "Hologic or similar Mo/Mo AGMS Mammo",
                        "profile": "AGMS2_STD_10.agp",
                        "sensor": "AGMS",
                        "order": 1,
                        "meta": {
                           "manufacturer": "Hologic",
                           "model": "M-II",
                           "anode": "Mo",
                           "filter": "Mo",
                           "doseOnly": false,
                           "kvMin": "21",
                           "kvMax": "49",
                           "filtrationMax": "44",
                           "hvlMin": "0.21",
                           "hvlMax": "0.6",
                           "paddle": true,
                           "tube": "General Mammo",
                           "conditions": "Mammography: Mo/Mo, Hologic or similar, 21-49 kV, 16-44 um Mo, with paddle",
                           "baseCal": "MoMo",
                           "baseCalId": "2",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-5",
                        "label": "General Mo/Rh AGMS Mammo",
                        "labelLong": "Hologic or similar Mo/Rh AGMS Mammo",
                        "profile": "AGMS3_STD_10.agp",
                        "sensor": "AGMS",
                        "order": 2,
                        "meta": {
                           "manufacturer": "Hologic",
                           "model": "M-II",
                           "anode": "Mo",
                           "filter": "Rh",
                           "doseOnly": false,
                           "kvMin": "21",
                           "kvMax": "49",
                           "filtrationMax": "42",
                           "hvlMin": "0.18",
                           "hvlMax": "0.56",
                           "paddle": true,
                           "tube": "General Mammo",
                           "conditions": "Mammography: Mo/Rh, Hologic or similar, 21-49 kV, 6-42 um Rh, with paddle",
                           "baseCal": "MoRh",
                           "baseCalId": "3",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-6",
                        "label": "General W/Ag AGMS Mammo",
                        "labelLong": "Hologic or similar W/Ag AGMS Mammo",
                        "profile": "AGMS7_STD_10.agp",
                        "sensor": "AGMS",
                        "order": 3,
                        "meta": {
                           "manufacturer": "Hologic",
                           "model": "M-IV",
                           "anode": "W",
                           "filter": "Ag",
                           "doseOnly": false,
                           "kvMin": "20",
                           "kvMax": "40",
                           "filtrationMax": "80",
                           "hvlMin": "0.17",
                           "hvlMax": "0.78",
                           "paddle": true,
                           "tube": "General Mammo",
                           "conditions": "Mammography: W/Ag, Hologic or similar, 21-40 kV, 12-80 um Ag, with paddle",
                           "baseCal": "Wag",
                           "baseCalId": "7",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-2",
                        "label": "General W/Rh AGMS Mammo",
                        "labelLong": "Hologic or similar W/Rh AGMS Mammo",
                        "profile": "AGMS8_STD_10.agp",
                        "sensor": "AGMS",
                        "order": 4,
                        "meta": {
                           "manufacturer": "Hologic",
                           "model": "M-IV",
                           "anode": "W",
                           "filter": "Rh",
                           "doseOnly": false,
                           "kvMin": "20",
                           "kvMax": "40",
                           "filtrationMax": "78",
                           "hvlMin": "0.17",
                           "hvlMax": "0.69",
                           "paddle": true,
                           "tube": "General Mammo",
                           "conditions": "Mammography: W/Rh, Hologic or similar, 20-40 kV, 12-78 um Rh, with paddle",
                           "baseCal": "WRh",
                           "baseCalId": "8",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-3",
                        "label": "General W/Al AGMS Mammo",
                        "labelLong": "Hologic or similar W/Al AGMS Mammo",
                        "profile": "AGMS9_STD_10.agp",
                        "sensor": "AGMS",
                        "order": 5,
                        "meta": {
                           "manufacturer": "Hologic",
                           "model": "M-IV",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "20",
                           "kvMax": "50",
                           "filtrationMax": "3030",
                           "hvlMin": "0.16",
                           "hvlMax": "1.96",
                           "paddle": true,
                           "tube": "General Mammo",
                           "conditions": "Mammography: W/Al, Hologic or similar, 20-50 kV, 300-3000 um Al, with paddle",
                           "baseCal": "WalMam",
                           "baseCalId": "9",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-9",
                  "label": "Multiple Sensors",
                  "order": "5",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-54",
                        "label": "DD Std trig + AGMS Diag",
                        "labelLong": "Dose Diode Std trigger + AGMS Diag",
                        "profile": "dd_std_agms1_10.agp",
                        "sensor": "DD/AGMS",
                        "order": 1,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: Use a dose diode to trigger an AGMS, W/Al, 40-150 kV, 2-40 mm Al ",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-55",
                        "label": "DD Low trig + AGMS Diag",
                        "labelLong": "Dose Diode Trig Low + AGMS Diag",
                        "profile": "dd_low_agms1_10.agp",
                        "sensor": "DD/AGMS",
                        "order": 2,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: For low dose rates, useing a dose diode to trigger an AGMS , W/Al,  40-150 kV, 2-40 mm Al ",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "LOW"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-42",
                        "label": "DD Std trig + Ich",
                        "labelLong": "Dose Diode Std trigger + Ion Chamber",
                        "profile": "DD_std_icf_10.agp",
                        "sensor": "DD/IC",
                        "order": 3,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "0",
                           "kvMax": "0",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "General Radiation: Use a dose diode to trigger low dose rate ion chamber measurments. ",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-57",
                        "label": "DD Low trig + Ich",
                        "labelLong": "Dose Diode Low Trig + Ion Chamber",
                        "profile": "dd_low_icf_10.agp",
                        "sensor": "DD/ICH",
                        "order": 4,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "conditions": "Diagnostic: For low dose rates, using a dose diode to trigger an ion chamber.",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "LOW"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-86",
                        "label": "AGMS Diag - ROI last 0.1s + Ich",
                        "labelLong": "AGMS Diag W/Al with ROI for last 100 ms + Ich",
                        "profile": "AGMS1_std_icf_adj0.1s_10.agp",
                        "sensor": "AGMS/ICH",
                        "order": 5,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "extractedRegion": "0.1 s",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: AGMS + ion chamber with 100 ms ROI at end of exposure, W/Al 40-160 kV, 2-30 mm Al",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-58",
                        "label": "AGMS Diag Std + mAs",
                        "labelLong": "AGMS Diag Std Trig + mAs sensor",
                        "profile": "AGMS1_std_ma_10.agp",
                        "sensor": "AGMS/MA",
                        "order": 6,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "conditions": "Diagnostic: AGMS plus mAs sensor, W/Al, 40-150 kV 2-40 mm Al",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-59",
                        "label": "ICh Std trigger + mAs",
                        "labelLong": "Ion Chamber Std Trig + mAs sensor",
                        "profile": "icf_std_ma_10.agp",
                        "sensor": "ICH/MA",
                        "order": 7,
                        "meta": {
                           "manufacturer": "General",
                           "doseOnly": false,
                           "kvMin": "20",
                           "kvMax": "150",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "conditions": "Diagnostic: Ion Chamber plus mAs sensor, 20-150 kV",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-7",
                  "label": "Dental",
                  "order": "6",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-30",
                        "label": "Pan Dental W/Al AGMS Diag",
                        "labelLong": "Pan Dental machines W/Al AGMS Diag",
                        "profile": "AGMS1_STD_PanDental_W-Al_10.agp",
                        "sensor": "AGMS",
                        "order": 1,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Panoramic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "60",
                           "kvMax": "100",
                           "filtrationMax": "4",
                           "hvlMin": "1.3",
                           "hvlMax": "5",
                           "paddle": false,
                           "tube": "Dental",
                           "conditions": "Pan Dental: W/Al, AGMS, 60-100 kV, 1.5-4 mm Al, End of exposure ROI = 1s, Std threshold",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-91",
                        "label": "Pan Dental AGMS w/ 10X60-DAP",
                        "labelLong": "Pan Dental machines W/Al AGMS Diag plus 10X60-DAP",
                        "profile": "agms1_std_PanDental_icfdap_10.agp",
                        "sensor": "AGMS/60DAP",
                        "order": 2,
                        "meta": {
                           "manufacturer": "Dental",
                           "model": "Panoramic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "60",
                           "kvMax": "100",
                           "filtrationMax": "4",
                           "hvlMin": "1.3",
                           "hvlMax": "5",
                           "paddle": false,
                           "tube": "Dental",
                           "conditions": "Pan Dental: W/Al, AGMS, 60-100 kV, 1.5-4 mm Al, End of exposure ROI = 1s, Std threshold, 10X6-60DAP chamber mounted on tube side",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-38",
                        "label": "Dental w/preheat AGMS Diag",
                        "labelLong": "Dental machines with preheat AGMS Diag",
                        "profile": "agms1_std_dental_10.agp",
                        "sensor": "AGMS",
                        "order": 3,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diag",
                           "doseOnly": false,
                           "kvMin": "60",
                           "kvMax": "100",
                           "filtrationMax": "4",
                           "hvlMin": "1.3",
                           "hvlMax": "5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Dental: W/Al, AGMS, with preheat e.g.Gendex 770, 60-100 kV, 1.5-4 mm Al. The time and pulse count are excluding the pre-heat region. KVp is calculated over the last 60% of main exposure. Dose is calculated for the entire exposure.",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-47",
                        "label": "Dental w/o preheat AGMS Diag",
                        "labelLong": "Dental Machine without preheat AGMS Diag",
                        "profile": "AGMS1_std_10.agp",
                        "sensor": "AGMS",
                        "order": 4,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "120",
                           "filtrationMax": "40",
                           "hvlMin": "1.3",
                           "hvlMax": "13.5",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Dental: W/Al,  AGMS, no preheat 40-120 kV, 2-40 mmAl, ",
                           "baseCal": "WAlDiag",
                           "baseCalId": "1",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-8",
                  "label": "DAP sensors",
                  "order": "7",
                  "items": [
                     {
                        "type": "profile",
                        "id": "p-35",
                        "label": "DAPCHK+",
                        "labelLong": "DAPCHK+ measurement",
                        "profile": "dapchk_std_10.agp",
                        "sensor": "DAPCHK+",
                        "order": 1,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "DAP: W/Al, DAPCHK+, 40-150 kV,  2-40 mm Al ",
                           "baseCal": "WalDiag",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-36",
                        "label": "10X6-60DAP",
                        "labelLong": "10X6-60DAP as DAP chamber to measure DAP",
                        "profile": "icf_std_dap_10.agp",
                        "sensor": "DAP",
                        "order": 2,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "DAP: 10X6-60DAP, Use chamber positioner & expose within the outline, 40-150 kV, 1.5-40 mm Al.",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-78",
                        "label": "10X6-60DAP as Ion Chamber",
                        "labelLong": "10X6-60DAP as Ion Chamber to measure Dose",
                        "profile": "icf_std_10.agp",
                        "sensor": "IC",
                        "order": 3,
                        "meta": {
                           "manufacturer": "Radcal",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "40",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: 10X6-60 DAP measure Dose.W/Al, 40-150 2-40 mm Al,  Expect wider energy response tolerance, +5%/-10%",
                           "baseCalId": "0",
                           "dateLastMod": "2024-01-17",
                           "threshold": "STD"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-93",
                        "label": "10X6-60DAP Free Run",
                        "labelLong": "10X6-60DAP Chamber Free Run",
                        "profile": "free_icf_dap.agp",
                        "sensor": "DAP",
                        "order": 4,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "doseOnly": false,
                           "filtrationMax": "40",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "Diagnostic: 10X6-60DAP Chamber Free run - manual trigger, W/Al, 40-150 kV 2-40 mm Al.",
                           "baseCalId": "0",
                           "dateLastMod": "2024-04-26",
                           "threshold": "FREE"
                        }
                     },
                     {
                        "type": "profile",
                        "id": "p-96",
                        "label": "10X6-60DAP with AGMS-Diag",
                        "labelLong": "10X6-60DAP with AGMS-Diag",
                        "profile": "icf_std_dap_agms1.agp",
                        "sensor": "IC/AGMS",
                        "order": 5,
                        "meta": {
                           "manufacturer": "General",
                           "model": "Diagnostic",
                           "anode": "W",
                           "filter": "Al",
                           "doseOnly": false,
                           "kvMin": "40",
                           "kvMax": "150",
                           "filtrationMax": "0",
                           "hvlMin": "0",
                           "hvlMax": "0",
                           "paddle": false,
                           "tube": "General",
                           "conditions": "DAP chamber with AGMS, IC trigger",
                           "baseCalId": "0",
                           "dateLastMod": "2024-09-10",
                           "threshold": "STD"
                        }
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-10",
                  "label": "Special Cases",
                  "order": "8",
                  "items": [
                     {
                        "type": "folder",
                        "id": "f-21",
                        "label": "Free Run",
                        "order": "1",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-62",
                              "label": "AGMS Diag Free Run",
                              "labelLong": "AGMS Diag Free Run",
                              "profile": "free_agms1.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "1.3",
                                 "hvlMax": "13.5",
                                 "paddle": true,
                                 "tube": "General",
                                 "conditions": "Diagnostic: AGMS Free run - manual trigger,  W/Al, 20-150 kV 2-40 mm Al.",
                                 "baseCal": "WalDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-63",
                              "label": "AGMS Diag + DD Free Run",
                              "labelLong": "AGMS Diag +  Dose Diode Free Run",
                              "profile": "free_agms1_dd.agp",
                              "sensor": "AGMS/DD",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "1.3",
                                 "hvlMax": "13.5",
                                 "paddle": true,
                                 "tube": "General",
                                 "conditions": "Diagnostic: AGMS+ Dose Diode Free run - manual trigger,  W/Al, 20-150 kV 2-40 mm Al.",
                                 "baseCal": "WalDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-67",
                              "label": "AGMS Diag + Ich Free Run",
                              "labelLong": "AGMS Diag + Ion Chamber Free Run",
                              "profile": "free_agms1_icf.agp",
                              "sensor": "AGMS/ICH",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "1.3",
                                 "hvlMax": "13.5",
                                 "paddle": true,
                                 "tube": "General",
                                 "conditions": "Diagnostic: AGMS+ Ion Chamber Free run - manual trigger,  W/Al, 20-150 kV 2-40 mm Al.",
                                 "baseCal": "WalDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-92",
                              "label": "DD Free Run",
                              "labelLong": "Dose Diode Free Run",
                              "profile": "free_dd.agp",
                              "sensor": "DD",
                              "order": 4,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "filtrationMax": "40",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: Dose Diode  Free run - manual trigger, W/Al, 40-150 kV 2-40 mm Al.",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-04-26",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-71",
                              "label": "DD + Ich Free Run",
                              "labelLong": "Dose Diode + Ion Chamber Free Run",
                              "profile": "free_dd_icf.agp",
                              "sensor": "DD/IC",
                              "order": 5,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "filtrationMax": "40",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: Dose Diode + Ion Chamber Free run - manual trigger, W/Al, 40-150 kV 2-40 mm Al.",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-73",
                              "label": "Ich Free Run",
                              "labelLong": "Ion Chamber Free Run",
                              "profile": "free_icf.agp",
                              "sensor": "IC",
                              "order": 6,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "filtrationMax": "40",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: Ion Chamber Free run - manual trigger, 20-150 kV ",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-93",
                              "label": "10X6-60DAP Free Run",
                              "labelLong": "10X6-60DAP Chamber Free Run",
                              "profile": "free_icf_dap.agp",
                              "sensor": "DAP",
                              "order": 7,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "filtrationMax": "40",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: 10X6-60DAP Chamber Free run - manual trigger, W/Al, 40-150 kV 2-40 mm Al.",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-04-26",
                                 "threshold": "FREE"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-22",
                        "label": "Ion Chamber",
                        "order": "2",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-80",
                              "label": "Ich Fast rise low trigger",
                              "labelLong": "Ion Chamber Low trigger for Fast rise time signals",
                              "profile": "icf_low_10.agp",
                              "sensor": "IC",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: W/Al Any ion chamber, low trigger & fast rise time",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "LOW"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-81",
                              "label": "Ich Slow rise low trigger",
                              "labelLong": "Ion Chamber - Low trigger, filtered",
                              "profile": "ics_low_10.agp",
                              "sensor": "IC",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: W/Al Any ion chamber, low trigger, filtered  (Slow rise time for improved noise rejection).",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "LOW"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-94",
                              "label": "Ich Slow rise std trigger",
                              "labelLong": "Ion Chamber - Std trigger, filtered",
                              "profile": "ics_std_10.agp",
                              "sensor": "IC",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: W/Al Any ion chamber, std trigger, filtered   (Slow rise time for improved noise rejection).",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-04-26",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-23",
                        "label": "2mm Filtration",
                        "order": "3",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-82",
                              "label": "Mo/Mo Dose Only +2mmAl -NP",
                              "labelLong": "AGMS Mo/Mo + 2mm Al - dose only - no paddle",
                              "profile": "agms4_stdDO_Mo-Mo2mm.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "anode": "Mo",
                                 "filter": "Mo",
                                 "doseOnly": true,
                                 "kvMin": "23",
                                 "kvMax": "30",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS Mo/Mo, Extra 2mm Al in beam, dose only, 23-30 kV,  No paddle",
                                 "baseCal": "DMR: Mo-Mo",
                                 "baseCalId": "4",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-84",
                              "label": "Rh/Ag Dose Only +2mmAl -NP",
                              "labelLong": "Rh/Ag + 2mm Al - dose only - no paddle AGMS Mammo",
                              "profile": "agms6_stdDO_Rh-Ag2mm.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "anode": "Rh",
                                 "filter": "Ag",
                                 "doseOnly": true,
                                 "kvMin": "27",
                                 "kvMax": "40",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, Rh/Ag, Extra 2mm Al in beam, dose only, 27-40 kV,  No paddle",
                                 "baseCal": "DMR: Rh-Rh",
                                 "baseCalId": "6",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-87",
                              "label": "W/Ag Dose Only +2mmAl 30kV",
                              "labelLong": "AGMS W/Ag + 2mm Al - dose only - no paddle 30kV",
                              "profile": "agms7_stdDO_W-Ag2mm_10.agp",
                              "sensor": "AGMS",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Ag",
                                 "doseOnly": true,
                                 "kvMin": "31",
                                 "kvMax": "34",
                                 "filtrationMax": "2000",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, W/Ag, Hologic, Extra 2mm Al in beam, dose only, 30 Kv,  No paddle",
                                 "baseCal": "Wag",
                                 "baseCalId": "7",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-90",
                              "label": "W/Rh Dose Only +2mmAl 29kV",
                              "labelLong": "AGMS W/Rh + 2mm Al - dose only - no paddle 29kV",
                              "profile": "agms8_stdDO_W-Rh2mmLow_10.agp",
                              "sensor": "AGMS",
                              "order": 4,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Rh",
                                 "doseOnly": true,
                                 "kvMin": "25",
                                 "kvMax": "29",
                                 "filtrationMax": "2000",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, W/Rh, Hologic, Extra 2mm Al in beam, dose only, 29V,  No paddle",
                                 "baseCal": "WRh",
                                 "baseCalId": "8",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-89",
                              "label": "W/Al Dose Only +2mmAl 32kV",
                              "labelLong": "AGMS W/Al + 2mm Al - dose only - no paddle 32kV",
                              "profile": "AGMS9_stdDO_2mm_10.agp",
                              "sensor": "AGMS",
                              "order": 5,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": true,
                                 "kvMin": "26",
                                 "kvMax": "38",
                                 "filtrationMax": "2000",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, W/Al, Hologic, Extra 2mm Al in beam, dose only, 32kV,  No paddle",
                                 "baseCal": "Wal",
                                 "baseCalId": "9",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     }
                  ]
               }
            ]
         },
         {
            "type": "category",
            "id": "c-2",
            "label": "Manufacturer",
            "order": "2",
            "items": [
               {
                  "type": "folder",
                  "id": "f-10",
                  "label": "GE",
                  "order": "1",
                  "items": [
                     {
                        "type": "folder",
                        "id": "f-21",
                        "label": "Pristina",
                        "order": "1",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-7",
                              "label": "Pristina Mo/Mo AGMS Mammo",
                              "labelLong": "GE Pristina Mo/Mo AGMS Mammo",
                              "profile": "AGMS4_STD_Pristina_Mo-Mo_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Mo",
                                 "filter": "Mo",
                                 "doseOnly": false,
                                 "kvMin": "22",
                                 "kvMax": "32",
                                 "filtrationMax": "44",
                                 "hvlMin": "0.24",
                                 "hvlMax": "0.51",
                                 "paddle": true,
                                 "tube": "GE Pristina",
                                 "conditions": "Mammography: Mo/Mo, GE Pristina, 22-32 kV, 16-44 um Mo, with paddle",
                                 "baseCal": "MoMoDMR",
                                 "baseCalId": "4",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-82",
                              "label": "Mo/Mo Dose Only +2mmAl -NP",
                              "labelLong": "AGMS Mo/Mo + 2mm Al - dose only - no paddle",
                              "profile": "agms4_stdDO_Mo-Mo2mm.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "anode": "Mo",
                                 "filter": "Mo",
                                 "doseOnly": true,
                                 "kvMin": "23",
                                 "kvMax": "30",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS Mo/Mo, Extra 2mm Al in beam, dose only, 23-30 kV,  No paddle",
                                 "baseCal": "DMR: Mo-Mo",
                                 "baseCalId": "4",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-8",
                              "label": "Pristina Rh/Ag AGMS Mammo",
                              "labelLong": "GE Pristina Rh/Ag AGMS Mammo",
                              "profile": "AGMS6_STD_Pristina_Rh-Ag_10.agp",
                              "sensor": "AGMS",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Rh",
                                 "filter": "Ag",
                                 "doseOnly": false,
                                 "kvMin": "27",
                                 "kvMax": "40",
                                 "filtrationMax": "80",
                                 "hvlMin": "0.27",
                                 "hvlMax": "0.78",
                                 "paddle": true,
                                 "tube": "GE Pristina",
                                 "conditions": "Mammography: Rh/Ag, GE Pristina, 27-40 kV, 12-80 um Mo with paddle",
                                 "baseCal": "RhRhDMR",
                                 "baseCalId": "6",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-84",
                              "label": "Rh/Ag Dose Only +2mmAl -NP",
                              "labelLong": "Rh/Ag + 2mm Al - dose only - no paddle AGMS Mammo",
                              "profile": "agms6_stdDO_Rh-Ag2mm.agp",
                              "sensor": "AGMS",
                              "order": 4,
                              "meta": {
                                 "anode": "Rh",
                                 "filter": "Ag",
                                 "doseOnly": true,
                                 "kvMin": "27",
                                 "kvMax": "40",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, Rh/Ag, Extra 2mm Al in beam, dose only, 27-40 kV,  No paddle",
                                 "baseCal": "DMR: Rh-Rh",
                                 "baseCalId": "6",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-9",
                              "label": "Pristina Rh/Ag IQST AGMS Mam",
                              "labelLong": "Pristina Rh/Ag with IQST procedure AGMS Mammo",
                              "profile": "AGMS10_STD_Pristina_IQST_10.agp",
                              "sensor": "AGMS",
                              "order": 5,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Rh",
                                 "filter": "Ag",
                                 "doseOnly": false,
                                 "kvMin": "34",
                                 "kvMax": "34",
                                 "paddle": false,
                                 "tube": "GE Pristina IQST",
                                 "conditions": "Mammography: Rh/Ag, GE Pristina IQST procedure, 34 kV only, no paddle",
                                 "baseCal": "RhCuDMR",
                                 "baseCalId": "10",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-10",
                              "label": "Pristina Mo/Cu AGMS Mammo",
                              "labelLong": "GE Pristina Mo/Cu AGMS",
                              "profile": "AGMS10_STD_Pristina_Mo-Cu_10.agp",
                              "sensor": "AGMS",
                              "order": 6,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Mo",
                                 "filter": "Cu",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "49",
                                 "filtrationMax": "250",
                                 "hvlMin": "2.48",
                                 "hvlMax": "3.27",
                                 "paddle": true,
                                 "tube": "GE Pristina",
                                 "conditions": "Mammography: Mo/Cu, GE Pristina, 40-49 kV, 250 um Cu, with paddle",
                                 "baseCal": "RhCuDMR",
                                 "baseCalId": "10",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-11",
                              "label": "Pristina Rh/Cu AGMS Mammo",
                              "labelLong": "GE Pristina Rh/Cu AGMS Mammo",
                              "profile": "agms10_std_Pristina_Rh-Cu_10.agp",
                              "sensor": "AGMS",
                              "order": 7,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Rh",
                                 "filter": "Cu",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "49",
                                 "filtrationMax": "250",
                                 "hvlMin": "2.42",
                                 "hvlMax": "3.2",
                                 "paddle": true,
                                 "tube": "GE Pristina",
                                 "conditions": "Mammography: Rh/Cu, GE Pristina, 40-49 kV, 250 um Cu, with paddle",
                                 "baseCal": "RhCuDMR",
                                 "baseCalId": "10",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-43",
                              "label": "Pristina Mo/Rh-Rh/Rh AGMS-M",
                              "labelLong": "Pristina Combo Mode with 2 regions of interest Mo/Rh-Rh/Rh",
                              "profile": "AGMS56_high_PristinaCombo_pp.agp",
                              "sensor": "AGMS",
                              "order": 8,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Rh",
                                 "filter": "Mo/Rh",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": true,
                                 "tube": "3D",
                                 "conditions": "Mammography: Rh/Mo,Rh Pristina Combo mode 2D + 3D  - provides  2ROI with Rh/Mo & Rh/Rh, with paddle",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "HIGH"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-44",
                              "label": "Pristina Mo/Rh-Rh/Ag 2 ROI",
                              "labelLong": "Pristina Combo Mode Mo-Rh/Rh-Ag AGMS Mammo - 2 regions of interest",
                              "profile": "AGMS56_high_PristinaAgCombo_pp.agp",
                              "sensor": "AGMS",
                              "order": 9,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Rh",
                                 "filter": "Mo/Ag",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": true,
                                 "tube": "3D",
                                 "conditions": "Mammography: Rh/Mo,Ag Pristina Combo mode 2D + 3D  - provides  2ROI with Rh/Mo & Rh/Ag, with paddle",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "HIGH"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-45",
                              "label": "Pristina Rh/Ag-Rh/Cu AGMS-M",
                              "labelLong": "Pristina Combo Mode with 2 regions of interest Rh/Ag-Rh/Cu",
                              "profile": "AGMS610_high_PristinaAgCuCombo_pp.agp",
                              "sensor": "AGMS",
                              "order": 10,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Pristina",
                                 "anode": "Rh",
                                 "filter": "Ag/Cu",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": true,
                                 "tube": "3D",
                                 "conditions": "Mammography: Rh/Ag,Cu Pristina Combo mode 2D & 3D  - provides  2ROI with Rh/Ag & Rh/Cu, with paddle",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "HIGH"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-22",
                        "label": "Essential",
                        "order": "2",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-12",
                              "label": "GE Essential Mo/Mo AGMS Mam",
                              "labelLong": "AGMS GE Essential Mo/Mo Mammo",
                              "profile": "AGMS4_STD_Essential_Mo-Mo_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Essential",
                                 "anode": "Mo",
                                 "filter": "Mo",
                                 "doseOnly": false,
                                 "kvMin": "22",
                                 "kvMax": "48",
                                 "filtrationMax": "44",
                                 "hvlMin": "0.24",
                                 "hvlMax": "0.51",
                                 "paddle": true,
                                 "tube": "GE Essential",
                                 "conditions": "Mammography: Mo/Mo, GE Essential 22-48 kV, 16-44 um Mo, with paddle",
                                 "baseCal": "MoMoDMR",
                                 "baseCalId": "4",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-13",
                              "label": "GE Essential Mo/Rh AGMS Mam",
                              "labelLong": "GE Essential Mo/Rh AGMS Mammo",
                              "profile": "AGMS5_STD_Essential_Mo-Rh_10.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Essential",
                                 "anode": "Mo",
                                 "filter": "Rh",
                                 "doseOnly": false,
                                 "kvMin": "22",
                                 "kvMax": "48",
                                 "filtrationMax": "42",
                                 "hvlMin": "0.2",
                                 "hvlMax": "0.56",
                                 "paddle": true,
                                 "tube": "GE Essential",
                                 "conditions": "Mammography: Mo/Rh, GE Essential 22-48 kV, 6-42 um Rh, with paddle",
                                 "baseCal": "MoRhDMR",
                                 "baseCalId": "5",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-14",
                              "label": "GE Essential Rh/Rh AGMS Mam",
                              "labelLong": "GE Essential Rh/Rh AGMS Mammo",
                              "profile": "AGMS6_STD_Essential_Rh-Rh_10.agp",
                              "sensor": "AGMS",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Essential",
                                 "anode": "Rh",
                                 "filter": "Rh",
                                 "doseOnly": false,
                                 "kvMin": "25",
                                 "kvMax": "48",
                                 "filtrationMax": "80",
                                 "hvlMin": "0.27",
                                 "hvlMax": "0.78",
                                 "paddle": true,
                                 "tube": "GE Essential",
                                 "conditions": "Mammography: Rh/Rh, GE Essential 25-48 kV, 12-80 um Rh, with paddle",
                                 "baseCal": "RhRhDMR",
                                 "baseCalId": "6",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-23",
                        "label": "DMR",
                        "order": "3",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-15",
                              "label": "GE DMR Mo/Mo AGMS Mammo",
                              "labelLong": "GE DMR or similar Mo/Mo AGMS Mammo",
                              "profile": "AGMS4_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "DMR",
                                 "anode": "Mo",
                                 "filter": "Mo",
                                 "doseOnly": false,
                                 "kvMin": "22",
                                 "kvMax": "48",
                                 "filtrationMax": "44",
                                 "hvlMin": "0.24",
                                 "hvlMax": "0.51",
                                 "paddle": true,
                                 "tube": "GE DMR",
                                 "conditions": "Mammography: Mo/Mo, GE DMR 22-48 kV, 16-44 um Mo, with paddle",
                                 "baseCal": "MoMoDMR",
                                 "baseCalId": "4",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-16",
                              "label": "GE DMR Mo/Rh AGMS Mammo",
                              "labelLong": "GE DMR or similar Mo/Rh AGMS Mammo",
                              "profile": "AGMS5_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "DMR",
                                 "anode": "Mo",
                                 "filter": "Rh",
                                 "doseOnly": false,
                                 "kvMin": "22",
                                 "kvMax": "48",
                                 "filtrationMax": "42",
                                 "hvlMin": "0.2",
                                 "hvlMax": "0.56",
                                 "paddle": true,
                                 "tube": "GE DMR",
                                 "conditions": "Mammography: Mo/Rh, GE DMR 22-48 kV, 6-42 um Rh, with paddle",
                                 "baseCal": "MoRhDMR",
                                 "baseCalId": "5",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-18",
                              "label": "GE DMR Mo/Al AGMS Mammo",
                              "labelLong": "GE DMR or similar Mo/Al AGMS Mammo",
                              "profile": "AGMS9_STD_DMR_Mo-Al_10.agp",
                              "sensor": "AGMS",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "DMR",
                                 "anode": "Mo",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "22",
                                 "kvMax": "49",
                                 "filtrationMax": "1000",
                                 "hvlMin": "0.42",
                                 "hvlMax": "0.61",
                                 "paddle": true,
                                 "tube": "GE DMR",
                                 "conditions": "Mammography: Mo/Al, GE DMR 22-49 kV, 1 mm Al, with paddle",
                                 "baseCal": "WAlMam",
                                 "baseCalId": "9",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-17",
                              "label": "GE DMR Rh/Rh AGMS Mammo",
                              "labelLong": "GE DMR or similar Rh/Rh AGMS Mammo",
                              "profile": "AGMS6_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 4,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "DMR",
                                 "anode": "Rh",
                                 "filter": "Rh",
                                 "doseOnly": false,
                                 "kvMin": "25",
                                 "kvMax": "48",
                                 "filtrationMax": "80",
                                 "hvlMin": "0.27",
                                 "hvlMax": "0.78",
                                 "paddle": true,
                                 "tube": "GE DMR",
                                 "conditions": "Mammography: Rh/Rh, GE DMR 25-48 kV, 12-480 um Rh, with paddle",
                                 "baseCal": "RhRhDMR",
                                 "baseCalId": "6",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-19",
                              "label": "GE DMR Rh/Al AGMS Mammo",
                              "labelLong": "GE DMR or similar Rh/Al AGMS Mammo",
                              "profile": "AGMS9_STD_DMR_Rh-Al_10.agp",
                              "sensor": "AGMS",
                              "order": 5,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "DMR",
                                 "anode": "Rh",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "25",
                                 "kvMax": "49",
                                 "filtrationMax": "1000",
                                 "hvlMin": "0.42",
                                 "hvlMax": "0.61",
                                 "paddle": true,
                                 "tube": "GE DMR",
                                 "conditions": "Mammography: Rh/Al, GE DMR 25-49 kV, 1 mm Al ,with paddle",
                                 "baseCal": "WAlMam",
                                 "baseCalId": "9",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-24",
                        "label": "Definium",
                        "order": "4",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-22",
                              "label": "GE Definium W/Al AGMS Diag",
                              "labelLong": "GE Definium W/Al AGMS Diag",
                              "profile": "AGMS1_STD_Definium_W-Al_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Definium",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "1.3",
                                 "hvlMax": "13.5",
                                 "paddle": false,
                                 "tube": "GE Definium (W)",
                                 "conditions": "Diagnostic: W/Al, GE Defimium 40-150 kV, 2-40 mm Al",
                                 "baseCal": "WAlDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-101",
                              "label": "GE Definium 656HD/Tempo DE",
                              "labelLong": "GE Definium 656HD/Tempo DE Subtraction AGMS Diag",
                              "profile": "GE_dual_energy_comboWAl_WAlDE.bm.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Definium 656HD/Tempo",
                                 "extractedRegion": "W/Al, W/Al (HE)",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "1.3",
                                 "hvlMax": "13.5",
                                 "paddle": false,
                                 "tube": "GE Definium (W)",
                                 "conditions": "Diagnostic: W/Al, GE Definium with Dual Energy Subtraction Mode 40-150 kV, 2-40 mm Al",
                                 "baseCal": "WAlDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2026-05-27",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-25",
                        "label": "Optima",
                        "order": "5",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-24",
                              "label": "GE Optima W-Al AGMS Diag",
                              "labelLong": "GE Optima W/Al AGMS Diag",
                              "profile": "AGMS1_STD_Optima_W-Al_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "GE",
                                 "model": "Optima",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "1.3",
                                 "hvlMax": "13.5",
                                 "paddle": false,
                                 "tube": "GE Optima (W)",
                                 "conditions": "Diagnostic: W/Al, GE Optima 40-150 kV, 2-40 mm Al",
                                 "baseCal": "WAlDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-11",
                  "label": "Hologic",
                  "order": "2",
                  "items": [
                     {
                        "type": "folder",
                        "id": "f-26",
                        "label": "Selenia Dimensions",
                        "order": "1",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-2",
                              "label": "General W/Rh AGMS Mammo",
                              "labelLong": "Hologic or similar W/Rh AGMS Mammo",
                              "profile": "AGMS8_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Rh",
                                 "doseOnly": false,
                                 "kvMin": "20",
                                 "kvMax": "40",
                                 "filtrationMax": "78",
                                 "hvlMin": "0.17",
                                 "hvlMax": "0.69",
                                 "paddle": true,
                                 "tube": "General Mammo",
                                 "conditions": "Mammography: W/Rh, Hologic or similar, 20-40 kV, 12-78 um Rh, with paddle",
                                 "baseCal": "WRh",
                                 "baseCalId": "8",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-6",
                              "label": "General W/Ag AGMS Mammo",
                              "labelLong": "Hologic or similar W/Ag AGMS Mammo",
                              "profile": "AGMS7_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Ag",
                                 "doseOnly": false,
                                 "kvMin": "20",
                                 "kvMax": "40",
                                 "filtrationMax": "80",
                                 "hvlMin": "0.17",
                                 "hvlMax": "0.78",
                                 "paddle": true,
                                 "tube": "General Mammo",
                                 "conditions": "Mammography: W/Ag, Hologic or similar, 21-40 kV, 12-80 um Ag, with paddle",
                                 "baseCal": "Wag",
                                 "baseCalId": "7",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-3",
                              "label": "General W/Al AGMS Mammo",
                              "labelLong": "Hologic or similar W/Al AGMS Mammo",
                              "profile": "AGMS9_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "20",
                                 "kvMax": "50",
                                 "filtrationMax": "3030",
                                 "hvlMin": "0.16",
                                 "hvlMax": "1.96",
                                 "paddle": true,
                                 "tube": "General Mammo",
                                 "conditions": "Mammography: W/Al, Hologic or similar, 20-50 kV, 300-3000 um Al, with paddle",
                                 "baseCal": "WalMam",
                                 "baseCalId": "9",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-28",
                              "label": "Hologic Dim W/Cu AGMS Mam",
                              "labelLong": "Hologic Dimensions W/Cu AGMS Mammo",
                              "profile": "AGMS10_STD_Dimensions_W-Cu_10.agp",
                              "sensor": "AGMS",
                              "order": 4,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "Dimension",
                                 "anode": "W",
                                 "filter": "Cu",
                                 "doseOnly": false,
                                 "kvMin": "38",
                                 "kvMax": "49",
                                 "filtrationMax": "325",
                                 "hvlMin": "1.54",
                                 "hvlMax": "3.52",
                                 "paddle": false,
                                 "tube": "Hologic Dimensions",
                                 "conditions": "Mammography: W/Cu, Hologic Dimensions, 38-49 kV, 174-325um Cu, no paddle",
                                 "baseCal": "RhCuDMR",
                                 "baseCalId": "10",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-32",
                              "label": "Hologic 2D/3D - W/Al-Rh-Cu",
                              "labelLong": "Hologic Combo Mode with 2 or 3 regions of Interest 2D or 3D - W/Al-W/Rh-W/Cu AGMS Mammo",
                              "profile": "AGMS9810_high_HologicCombo_pp.agp",
                              "sensor": "AGMS",
                              "order": 5,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "Dimensions",
                                 "anode": "W",
                                 "filter": "Al/Rh/Cu",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "2D/3D",
                                 "conditions": "Mammography: W/Al, Rh, Cu, Hologic Selenia Combo mode 2D + 3D, - provides  2 ROI with W/Al & W/Rh or 3 ROI with Add W/Cu, no paddle",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "HIGH"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-39",
                              "label": "Hologic 2D/3D - W/Al-Ag-Cu",
                              "labelLong": "Hologic Combo Mode with 2 or 3 regions of Interest 2D or 3D - W/Al-W/Ag-W/Cu AGMS Mammo",
                              "profile": "AGMS9710_high_HologicCombo_pp.agp",
                              "sensor": "AGMS",
                              "order": 6,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "Dimensions",
                                 "anode": "W",
                                 "filter": "Al/Ag/Cu",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "2D/3D",
                                 "conditions": "Mammography: W/Al, Ag, Cu, Hologic Selenia Combo mode 2D + 3D - provides 2 ROI with W/Al & W/Ag or 3 ROI Add W/Cu, no paddle",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "HIGH"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-87",
                              "label": "W/Ag Dose Only +2mmAl 30kV",
                              "labelLong": "AGMS W/Ag + 2mm Al - dose only - no paddle 30kV",
                              "profile": "agms7_stdDO_W-Ag2mm_10.agp",
                              "sensor": "AGMS",
                              "order": 7,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Ag",
                                 "doseOnly": true,
                                 "kvMin": "31",
                                 "kvMax": "34",
                                 "filtrationMax": "2000",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, W/Ag, Hologic, Extra 2mm Al in beam, dose only, 30 Kv,  No paddle",
                                 "baseCal": "Wag",
                                 "baseCalId": "7",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-90",
                              "label": "W/Rh Dose Only +2mmAl 29kV",
                              "labelLong": "AGMS W/Rh + 2mm Al - dose only - no paddle 29kV",
                              "profile": "agms8_stdDO_W-Rh2mmLow_10.agp",
                              "sensor": "AGMS",
                              "order": 8,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Rh",
                                 "doseOnly": true,
                                 "kvMin": "25",
                                 "kvMax": "29",
                                 "filtrationMax": "2000",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, W/Rh, Hologic, Extra 2mm Al in beam, dose only, 29V,  No paddle",
                                 "baseCal": "WRh",
                                 "baseCalId": "8",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-89",
                              "label": "W/Al Dose Only +2mmAl 32kV",
                              "labelLong": "AGMS W/Al + 2mm Al - dose only - no paddle 32kV",
                              "profile": "AGMS9_stdDO_2mm_10.agp",
                              "sensor": "AGMS",
                              "order": 9,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-IV",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": true,
                                 "kvMin": "26",
                                 "kvMax": "38",
                                 "filtrationMax": "2000",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Mammographic: AGMS, W/Al, Hologic, Extra 2mm Al in beam, dose only, 32kV,  No paddle",
                                 "baseCal": "Wal",
                                 "baseCalId": "9",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-28",
                        "label": "Selenia (Mo)",
                        "order": "2",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-4",
                              "label": "General Mo/Mo AGMS Mammo",
                              "labelLong": "Hologic or similar Mo/Mo AGMS Mammo",
                              "profile": "AGMS2_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-II",
                                 "anode": "Mo",
                                 "filter": "Mo",
                                 "doseOnly": false,
                                 "kvMin": "21",
                                 "kvMax": "49",
                                 "filtrationMax": "44",
                                 "hvlMin": "0.21",
                                 "hvlMax": "0.6",
                                 "paddle": true,
                                 "tube": "General Mammo",
                                 "conditions": "Mammography: Mo/Mo, Hologic or similar, 21-49 kV, 16-44 um Mo, with paddle",
                                 "baseCal": "MoMo",
                                 "baseCalId": "2",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-5",
                              "label": "General Mo/Rh AGMS Mammo",
                              "labelLong": "Hologic or similar Mo/Rh AGMS Mammo",
                              "profile": "AGMS3_STD_10.agp",
                              "sensor": "AGMS",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "Hologic",
                                 "model": "M-II",
                                 "anode": "Mo",
                                 "filter": "Rh",
                                 "doseOnly": false,
                                 "kvMin": "21",
                                 "kvMax": "49",
                                 "filtrationMax": "42",
                                 "hvlMin": "0.18",
                                 "hvlMax": "0.56",
                                 "paddle": true,
                                 "tube": "General Mammo",
                                 "conditions": "Mammography: Mo/Rh, Hologic or similar, 21-49 kV, 6-42 um Rh, with paddle",
                                 "baseCal": "MoRh",
                                 "baseCalId": "3",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-12",
                  "label": "Phillips",
                  "order": "3",
                  "items": [
                     {
                        "type": "folder",
                        "id": "f-29",
                        "label": "MicroDose",
                        "order": "1",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-25",
                              "label": "MicroDose W/Al AGMS Mammo",
                              "labelLong": "Philips MicroDose W/Al AGMS Mammol",
                              "profile": "AGMS9_STD_Microdose_W-Al_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Phillips",
                                 "model": "Microdose",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "20",
                                 "kvMax": "50",
                                 "filtrationMax": "3030",
                                 "hvlMin": "0.16",
                                 "hvlMax": "1.96",
                                 "paddle": false,
                                 "tube": "Phillips Microdose",
                                 "conditions": "Mammography: W/Al, Philips Microdose 20-50 kV, 0.3-3mm Al, no paddle",
                                 "baseCal": "WalMam",
                                 "baseCalId": "9",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-13",
                  "label": "Siemens",
                  "order": "4",
                  "items": [
                     {
                        "type": "folder",
                        "id": "f-30",
                        "label": "Mammomat Revelation",
                        "order": "1",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-27",
                              "label": "Revelation W/Ti AGMS Diag",
                              "labelLong": "Siemens Mammomat Revelation W-Ti AGMS Diag",
                              "profile": "AGMS1_STD_Mammomat_W-Ti_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Siemens",
                                 "model": "Mammomat Revelation",
                                 "anode": "W",
                                 "filter": "Ti",
                                 "doseOnly": false,
                                 "kvMin": "41",
                                 "kvMax": "50",
                                 "filtrationMax": "1",
                                 "hvlMin": "2.45",
                                 "hvlMax": "3",
                                 "paddle": true,
                                 "tube": "Mammomat Revelation",
                                 "conditions": "Mammography: W/Ti, Siemens Mammomat Revelation, 41-50 kV, 1mm Ti, no paddle",
                                 "baseCal": "WalDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-32",
                        "label": "Straton CT",
                        "order": "2",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-26",
                              "label": "Straton AGMS-D w/W kV filter",
                              "labelLong": "Siemens Straton CT W/Al AGMS Diag with Tungsten filter Model 8255",
                              "profile": "AGMS1_STD_StratonCT_W-Al_10.agp",
                              "sensor": "AGMS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Siemens",
                                 "model": "Straton CT",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "140",
                                 "hvlMin": "5",
                                 "hvlMax": "10",
                                 "paddle": false,
                                 "tube": "Straton CT (use Radcal 8155 filter)",
                                 "conditions": "CT: W/Al, Siemens Straton, W/Al 40-140 kV Radcal 8155 filter required",
                                 "baseCal": "WAlDiag",
                                 "baseCalId": "1",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     }
                  ]
               },
               {
                  "type": "folder",
                  "id": "f-9",
                  "label": "Radcal",
                  "order": "5",
                  "items": [
                     {
                        "type": "folder",
                        "id": "f-17",
                        "label": "Ion Chambers",
                        "order": "1",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-78",
                              "label": "10X6-60DAP as Ion Chamber",
                              "labelLong": "10X6-60DAP as Ion Chamber to measure Dose",
                              "profile": "icf_std_10.agp",
                              "sensor": "IC",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: 10X6-60 DAP measure Dose.W/Al, 40-150 2-40 mm Al,  Expect wider energy response tolerance, +5%/-10%",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-79",
                              "label": "Ich zero once for surveying",
                              "labelLong": "10X6-1800 used for surveying - low trigger zero once",
                              "profile": "icf_low_nrz_10.agp",
                              "sensor": "IC",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: W/Al Any ion chamber, like a 10X6-1800, low threshold, zero only once",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "LOW"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-16",
                        "label": "DAP Chamber",
                        "order": "2",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-36",
                              "label": "10X6-60DAP",
                              "labelLong": "10X6-60DAP as DAP chamber to measure DAP",
                              "profile": "icf_std_dap_10.agp",
                              "sensor": "DAP",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "DAP: 10X6-60DAP, Use chamber positioner & expose within the outline, 40-150 kV, 1.5-40 mm Al.",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-78",
                              "label": "10X6-60DAP as Ion Chamber",
                              "labelLong": "10X6-60DAP as Ion Chamber to measure Dose",
                              "profile": "icf_std_10.agp",
                              "sensor": "IC",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "40",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: 10X6-60 DAP measure Dose.W/Al, 40-150 2-40 mm Al,  Expect wider energy response tolerance, +5%/-10%",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-93",
                              "label": "10X6-60DAP Free Run",
                              "labelLong": "10X6-60DAP Chamber Free Run",
                              "profile": "free_icf_dap.agp",
                              "sensor": "DAP",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "filtrationMax": "40",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "Diagnostic: 10X6-60DAP Chamber Free run - manual trigger, W/Al, 40-150 kV 2-40 mm Al.",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-04-26",
                                 "threshold": "FREE"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-96",
                              "label": "10X6-60DAP with AGMS-Diag",
                              "labelLong": "10X6-60DAP with AGMS-Diag",
                              "profile": "icf_std_dap_agms1.agp",
                              "sensor": "IC/AGMS",
                              "order": 4,
                              "meta": {
                                 "manufacturer": "General",
                                 "model": "Diagnostic",
                                 "anode": "W",
                                 "filter": "Al",
                                 "doseOnly": false,
                                 "kvMin": "40",
                                 "kvMax": "150",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "tube": "General",
                                 "conditions": "DAP chamber with AGMS, IC trigger",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-09-10",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-15",
                        "label": "Light Sensor",
                        "order": "3",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-50",
                              "label": "AGLS",
                              "labelLong": "Accu-Gold Light Sensor",
                              "profile": "AGLS_high_10.agp",
                              "sensor": "AGLS",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "AG",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "0",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Light Sensor: AGLS Luminance and Illuminance",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "HIGH"
                              }
                           }
                        ]
                     },
                     {
                        "type": "folder",
                        "id": "f-14",
                        "label": "Dynalyzer",
                        "order": "4",
                        "items": [
                           {
                              "type": "profile",
                              "id": "p-48",
                              "label": "Dyn IV Std trigger",
                              "labelLong": "Dynalyzer IV Std trigger",
                              "profile": "dyn_std_10.agp",
                              "sensor": "DYN",
                              "order": 1,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "150",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Dynalyzer IV, 0-150 kV, Standard Trigger",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-51",
                              "label": "Dyn IV Std trigger + mA",
                              "labelLong": "Dynalyzer IV Std trigger + mA sensor",
                              "profile": "dyn_std_ma_10.agp",
                              "sensor": "DYN/MA",
                              "order": 2,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "150",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Dynalyzer IV + mA sensor, 0-150 kV, Standard Trigger",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           },
                           {
                              "type": "profile",
                              "id": "p-52",
                              "label": "Dyn IV Std trigger + mA + Ich",
                              "labelLong": "Dynalyzer IV Std trigger + mA sensor + ion chamber",
                              "profile": "dyn_std_icf_ma_10.agp",
                              "sensor": "DYN/MA/ICH",
                              "order": 3,
                              "meta": {
                                 "manufacturer": "Radcal",
                                 "model": "Diagnostic",
                                 "doseOnly": false,
                                 "kvMin": "0",
                                 "kvMax": "150",
                                 "filtrationMax": "0",
                                 "hvlMin": "0",
                                 "hvlMax": "0",
                                 "paddle": false,
                                 "conditions": "Dynalyzer IV + mA sensor + ion chamber, 20-150 kV, Standard Trigger",
                                 "baseCalId": "0",
                                 "dateLastMod": "2024-01-17",
                                 "threshold": "STD"
                              }
                           }
                        ]
                     }
                  ]
               }
            ]
         }
      ]
   }
}